Doctorbass wrote:Someone should explain how to measure the rds-on of both true and counterfait mosfet... I guess it is like a shunt work using 4 wire sensing..
i'll try that.. by activating it.. anyway i'll remove all the FB4110 on the crystalyte and put brand new true IR mosfet.
methods wrote:If anyone would like to take the time to characterize these (possibly) counterfeit mosfets I will send you a complementary set of REAL 4110's for your trouble free of charge
VGS = 10V, ID = 75A Pulse width Ã¢â€°Â¤ 400ÃŽÂ¼s; duty cycle Ã¢â€°Â¤ 2%.
liveforphysics wrote:Link- you can't just throw a meter across them to test them... You have to dynamicly load test them as you would test a battery Ri...
Link wrote:I don't exactly have the most precise equipment
Jeremy Harris wrote:I'm happy to be proved wrong, if someone can provide some evidence.
wrobinson0413 wrote: Sorry for the long posting...should have pruned the PCN reports a bit instead of just copying and pasting.
Jeremy Harris wrote: Why go to the bother of setting up a multi-million dollar bootleg fab plant and then make an error that is so obvious?
Users browsing this forum: Bing [Bot], Google [Bot], HypnoToad and 5 guests